The 05103-45 Wind Monitor is a lightweight instrument that measures wind speed and wind direction. It is designed to prevent ice buildup, allowing the sensor to provide accurate measurements in harsh alpine conditions. Manufactured by R. M. Young, this wind monitor is cabled for use with Campbell Scientific data loggers.Read More
|Operating Temperature Range||-50° to +50°C (assuming non-riming conditions)|
|Mounting Pipe Description||
|Propeller Diameter||14 cm (5.5 in.)|
|Overall Height||37 cm (14.6 in.)|
|Main Housing Diameter||5 cm (2.0 in.)|
|Overall Length||55 cm (21.7 in.)|
|Weight||1 kg (2.2 lb)|
|Range||0 to 100 m/s (0 to 224 mph)|
|Accuracy||±0.3 m/s (0.6 mph) or 1% of reading|
|Starting Threshold||1.0 m/s (2.2 mph)|
|Distance Constant||2.7 m (8.9 ft) 63% recovery|
ac voltage (three pulses per revolution)
90 Hz (1800 rpm) = 8.8 m/s (19.7 mph)
|Resolution||(0.0980 m s-1)/(scan rate in seconds) or (0.2192 mph)/(scan rate seconds)|
|Mechanical Range||0 to 360°|
|Electrical Range||355° (5° open)|
|Starting Threshold||1.1 m/s (2.4 mph) at 10° displacement|
|Distance Constant||1.3 m (4.3 ft) 50% recovery|
|Damped Natural Wavelength||7.4 m (24.3 ft)|
|Undamped Natural Wavelength||7.2 m (23.6 ft)|
|Voltage||Power switched excitation voltage supplied by data logger|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
The 05103-45 can be attached to a crossarm via a 17953 NU-RAIL fitting or CM220 Right Angle Mounting Bracket. Alternatively, it can be attached to the top of a CM106B or stainless-steel tripod via the CM216 Sensor Mounting Kit.
Wind profile studies measure many wind sensors. For these applications, the LLAC4 4-Channel Low Level AC Conversion Module can be used to increase the number of Wind Monitors measured by one data logger. The LLAC4 allows data logger control ports to read the wind speed sensor’s ac signals instead of using pulse channels. Data loggers compatible with the LLAC4 are the CR200(X) series (ac signal ≤1 kHz only), CR800, CR850, CR1000, CR3000, and CR5000.
The 05103-45's propeller uses one pulse count channel on the data logger. Its wind vane requires one single-ended channel and access to an excitation channel (the excitation channel can be shared with other high impedance sensors).
The 05103-45's propeller is measured by the PulseCount Instruction in CRBasic and by Instruction 3 (Pulse Count) in Edlog. The wind vane is measured by the BrHalf Instruction in CRBasic and by Instruction 4 (Excite-Delay-SE) in Edlog. The measurements are typically processed for output with the Wind Vector instruction (not present in the CR500 or CR9000 but is present in the CR9000X).
Number of FAQs related to 05103AP-L: 11
Expand AllCollapse All
This depends on what is broken. Typically, Campbell Scientific can repair the unit, and the user does not have to purchase a new one.
The measurement instructions will likely remain the same. However, in addition to the multiplier and offset, the type of pulse may change for the wind speed, and the excitation voltage may change for the wind direction. For an explanation of how the data logger needs to be programmed, see the instruction manual.
The short answer is less than 0.01 mA. The wind speed signal requires no power. The wind direction portion of the sensor only uses a maximum of 0.5 mA when excited with 5 Vdc, and then it is only on for 0.016 s for every measurement. When the wind direction is measured every second (typical), the average current drain is less than 0.01 mA.
Orientation of the wind monitor is done after the data logger has been programmed, and the location of True North has been determined. True North is usually found by reading a magnetic compass and applying the correction for magnetic declination, where magnetic declination is the number of degrees between True North and Magnetic North. Magnetic declination for a specific site can be obtained from a USFA map, local airport, or through a computer service.
Most Campbell Scientific sensors are available as an –L, which indicates a user-specified cable length. If a sensor is listed as an –LX model (where “X” is some other character), that sensor’s cable has a user-specified length, but it terminates with a specific connector for a unique system:
If a sensor does not have an –L or other –LX designation after the main model number, the sensor has a set cable length. The cable length is listed at the end of the Description field in the product’s Ordering information. For example, the 034B-ET model has a description of “Met One Wind Set for ET Station, 67 inch Cable.” Products with a set cable length terminate, as a default, with pigtails.
If a cable terminates with a special connector for a unique system, the end of the model number designates which system. For example, the 034B-ET model designates the sensor as a 034B for an ET107 system.
Not every sensor has different cable termination options. The options available for a particular sensor can be checked by looking in two places in the Ordering information area of the sensor product page:
If a sensor is offered in an –ET, –ETM, –LC, –LQ, or –QD version, that option’s availability is reflected in the sensor model number. For example, the 034B is offered as the 034B-ET, 034B-ETM, 034B-LC, 034B-LQ, and 034B-QD.
All of the other cable termination options, if available, are listed on the Ordering information area of the sensor product page under “Cable Termination Options.” For example, the 034B-L Wind Set is offered with the –CWS, –PT, and –PW options, as shown in the Ordering information area of the 034B-L product page.
Note: As newer products are added to our inventory, typically, we will list multiple cable termination options under a single sensor model rather than creating multiple model numbers. For example, the HC2S3-L has a –C cable termination option for connecting it to a CS110 instead of offering an HC2S3-LC model.
Many Campbell Scientific sensors are available with different cable termination options. These options include the following:
Note: The availability of cable termination options varies by sensor. For example, sensors may have none, two, or several options to choose from. If a desired option is not listed for a specific sensor, contact Campbell Scientific for assistance.