|-NOTE-||Note: Additional specifications are listed in the 21X Specifications Sheet.|
|Analog Input Channels||8 differential (16 single-ended)|
|Switched Excitation Outputs||4|
|Continuous Analog Outputs||2|
|Pulse Count Channels||4|
|Memory||Stores 19,296 data points in internal memory.|
A typical field-based 21X system consisted of:
• 21X Micrologger
• Weatherproof Enclosure
• Communications Peripheral(s)
• Programming and Communications Software
Number of FAQs related to 21X: 6
Expand AllCollapse All
A practical maximum is to connect one multiplexer per every two control terminals on the data logger. Control terminals can be shared between multiplexers to increase the number of connected multiplexers. Sharing terminals, however, requires more complex wiring and programming. Users who would like to connect more than one multiplexer per every two control terminals are advised to contact a sales or support engineer at Campbell Scientific for assistance.
It may be that there is no output flag set to dump data to final storage. It is possible that access is not being made to the right input location of the barometric pressure sensor to send it to final storage. To set an output flag, refer to the "Program Control Instructions" section in the 21X Micrologger Operator's Manual.
After the power to the 21X is turned off, the program, final storage data, and clock settings are lost. There is no battery-backup RAM or Clock in the 21X.
There is no built-in security code in the 21X. A security code can be set in one of two ways in the 21X:
For more information on security, refer to Page 1-6 of the 21X Operator's Manual.
Whenever power is lost to the 21X, the program, clock, and final storage data are lost. Using our data logger support software, the program can be written and reloaded to the 21X.
Campbell Scientific recommends keeping a backup to the program that includes a software and hard copy, as well as a backup wiring diagram of the sensors.