The 27106T Vertical Propeller Anemometer, manufactured by R. M. Young, is a low threshold precision air velocity sensor that is especially suited for your applications where you need to monitor the vertical wind component. It connects directly to a Campbell Scientific data logger, which measures the 27106T's signal and converts the signal to engineering units (mph, m/s, knots).Read More
|-NOTE-||Threshold and distance constant values are for axial flow.|
|Threshold Sensitivity||0.4 m s-1 (0.8 mph)|
|Distance Constant||< 2.1 m (6.9 ft)|
|Pitch||30.0 cm (11.8 in.) air passage per revolution|
Analog dc voltage proportional to axial wind component. Polarity reverses with reverse rotation.
1800 rpm (500 mV) = 9.0 m/s (20.1 mph)
|Operating Temperature Range||-50° to +50°C|
|Propeller Description||4-blade helicoid propeller molded of carbon fiber thermoplastic|
|Propeller Diameter||20 cm (8 in.)|
|Housing Diameter||2.5 cm (1 in.)|
|Length||43 cm (17 in.)|
|Weight||0.5 kg (1.2 lb)|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
The 27106T is shipped with a 3/4 inch IPS threaded pipe for mounting. The pipe can be attached to a CM202, CM202SS, CM203, CM204, CM204SS, or CM206 crossarm using a 1049 Nu-Rail fitting or CM220 Right-Angle Mounting Bracket.
The 27106T includes a connector/coupler that mates to the sensor and to the 3/4-inch IPS threaded pipe. The customer specified length of cable is soldered to the pins on the connector and the cable threads through the coupler and out the pipe.
Number of FAQs related to 27106T-L: 6
Expand AllCollapse All
Campbell Scientific does not currently manufacture or stock this item. The 27106T-L is shipped from R. M. Young with the mounting pipe attached. Contact R. M. Young directly for assistance.
For this information, refer to the “Wind Speed Multiplier and Offset” section in the 27106T instruction manual.
Most Campbell Scientific sensors are available as an –L, which indicates a user-specified cable length. If a sensor is listed as an –LX model (where “X” is some other character), that sensor’s cable has a user-specified length, but it terminates with a specific connector for a unique system:
If a sensor does not have an –L or other –LX designation after the main model number, the sensor has a set cable length. The cable length is listed at the end of the Description field in the product’s Ordering information. For example, the 034B-ET model has a description of “Met One Wind Set for ET Station, 67 inch Cable.” Products with a set cable length terminate, as a default, with pigtails.
If a cable terminates with a special connector for a unique system, the end of the model number designates which system. For example, the 034B-ET model designates the sensor as a 034B for an ET107 system.
Not every sensor has different cable termination options. The options available for a particular sensor can be checked by looking in two places in the Ordering information area of the sensor product page:
If a sensor is offered in an –ET, –ETM, –LC, –LQ, or –QD version, that option’s availability is reflected in the sensor model number. For example, the 034B is offered as the 034B-ET, 034B-ETM, 034B-LC, 034B-LQ, and 034B-QD.
All of the other cable termination options, if available, are listed on the Ordering information area of the sensor product page under “Cable Termination Options.” For example, the 034B-L Wind Set is offered with the –CWS, –PT, and –PW options, as shown in the Ordering information area of the 034B-L product page.
Note: As newer products are added to our inventory, typically, we will list multiple cable termination options under a single sensor model rather than creating multiple model numbers. For example, the HC2S3-L has a –C cable termination option for connecting it to a CS110 instead of offering an HC2S3-LC model.
The anemometer uses a single-ended voltage measurement.
Many Campbell Scientific sensors are available with different cable termination options. These options include the following:
Note: The availability of cable termination options varies by sensor. For example, sensors may have none, two, or several options to choose from. If a desired option is not listed for a specific sensor, contact Campbell Scientific for assistance.