The CR9000X is a large, modular multiprocessor system that provides precision measurement capabilities in a rugged, battery-operated package. It consists of a base system and a chassis with slots for up to nine user-selected I/O modules. The CR9000X is our fastest data logger, with a measurement rate of 100,000 Hz, making it ideal for rapid sampling applications.
Campbell Scientific also offers the CR9000XC, a compact version, that accepts up to five I/O modules. For more information, visit the CR9000XC Product Info page.Read More
|Operating Temperature Range||
|Analog Inputs||28 single-ended or 14 differential per CR9050, CR9051E, or CR9055(E) module|
|Pulse Counters||12 per CR9071 module|
|Switched 12 Volt||1 terminal|
|Analog Voltage Accuracy||±(0.07% of reading + 4 A/D counts), -25° to +50°C|
|Power Requirements||9.6 to 16 Vdc|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
Customers can add CR9000X dataloggers to networks containing the older CR9000 or CR9000C dataloggers. I/O modules other than the CR9080 can be used with either the CR9000 or CR9000X. CR9000 communication interfaces (i.e., NL105, BLC100, TL925, PLA100) are not compatible with the CR9000X, and therefore have been retired.
With several channel types, the CR9000X is compatible with many sensors, including thermocouples and 4 to 20 mA sensors.
Measurement and control peripherals typically used with the CR9000X are our AM25T 25-Channel Solid State Multiplexer, SDM-CAN Interface, SDM-INT8 Eight Channel Interval Timer, and SDM-SIO4 Serial Input/Output Module. Other measurement and control peripherals are compatible but they do not support the CR9000X datalogger's maximum measurement rate and are therefore impractical for most CR9000X applications.
The CR9000X typically communicates with a PC via direct connect or Ethernet. Because the CR9000X has an on-board 10baseT/100baseT port, an Ethernet interface such as the NL201 is not required.
Storage capacity can be increased by using a PC or CompactFlash card. The CR9000X's PCMCIA card slot supports one Type I, Type II, or Type III PC Card or the CF1 adapter and one CompactFlash (CF) card.
The storage capacity of Type II cards exceeds 1 GB. Type III cards provide data storage capacities exceeding 1 GB but may not be suitable for all environments. Campbell Scientific offers several CF cards that have passed our ESD testing and operate properly with our data loggers (see Ordering tab). Please note that the PCMCIA and CompactFlash cards need to be industrial-grade and have a storage capacity of 2 GB or less.
Other communication peripherals are compatible but they do not support the CR9000X datalogger's maximum measurement rate and are therefore impractical for most CR9000X applications.
Two enclosures are offered for the CR9000X. The 8253 fiberglass environmental enclosure is designed for field applications where the enclosure will be exposed to the elements. The 8255 lab enclosure is for applications where the CR9000X will reside inside a building.
CRBasic, the CR9000X's full programming language, supports simple or complex programming and many on-board data reduction processes. CRBasic is included in RTDAQ, LoggerNet, and PC400.
RTDAQ Real-Time Data Acquisition Software must be ordered separately; the CR9000X is also compatible with other Campbell Scientific software.
Current Operating System, Compiler and CR9000X support files for the CRBasic Editor. Requires the Device Configuration Utility, LoggerNet or RTDAQ to upload.
Use of this file will update the datalogger support files for the CRBasic Editor included in LoggerNet and RTDAQ.
Upgrade PC9000 version 5.0, 5.1, 5.2 or 5.3 to 5.3.1; no intermediate steps are required.
PC9000 5.0, 5.1, 5.2 or 5.3 must be installed on your machine.
Number of FAQs related to CR9000X: 46
Expand AllCollapse All
The CR9000X and CR9000XC differ only in the number of I/O cards they can hold. The CR9000X can hold 9 I/O cards, and the CR9000XC can hold 5 I/O cards. We provide both sizes to accommodate our customers; the same I/O cards can be used in either chassis.
There are two ways to upgrade an operating system:
Yes. The simplest method is to use conditional program statements that execute most of the code based on time. For example, the data could be scheduled to log at 6 a.m. and finish at 8 p.m. using CRBasic instructions such as TimeIntoInterval(). Another option is to use an IfThen/EndIf construction that does a logical test of light-level measurements based on a light sensor. An additional option is to use calculated sunrise and sunset times along with a combination of RealTime() and Case instructions.
For more information, see the “Decisions, Decisions, Decisions…” article.
The maximum cable length depends on the interface being used.
Common causes include the following:
The data logger will not reset the SW12 unless it is done under program control using the SW12() or PortSet() instructions, or unless the data logger compiles or recompiles a program. This could be done when a new program is sent to the data logger, or if the power is cycled.
The available COM ports listed in the COM Port drop-down menu are supplied to PC400 by the Windows Operating System (OS). If there are no COM ports shown for selection, it most likely means that there are no COM ports registered with the Windows OS. This can be confirmed using the Windows Device Manager (Control Panel | Device Manager | Ports).
Most modern laptops are not equipped with native RS-232 COM ports. In this situation, a USB-to-RS-232 adapter cable must be used to connect to the data logger. Even when the drivers for this device have been properly installed, the derived COM port will not be shown for selection until the cable is attached to the laptop.