The NL120 allows the data logger to communicate over a local network or a dedicated Internet connection via TCP/IP. This small, rugged communication device connects directly to the 40-pin peripheral port on a CR1000 or CR3000 datalogger.Read More
|Power Requirements||12 V (supplied through the data logger’s peripheral port)|
|Ethernet Speed||10Base-T (10 Mbps), half-duplex|
|Typical Current Drain||20 mA|
|Dimensions||10.2 x 6.4 x 2.8 cm (4.0 x 2.5 x 1.1 in.)|
|Weight||66.62 g (2.35 oz)|
EMI and ESD Protection
|European Standards||Meets requirements for a class A device under European Standards.|
|Application of Council Directive(s)||89/336/EEC as amended by 89/336/EEC and 93/68/EEC|
|Standards to Which Conformity Is Declared||EN55022-1; 1995 and EN50082-1: 1992|
|LoggerNet||Version 3.2 or later|
|PC400||Version 1.3 or later|
|DevConfig||Version 1.5 or later|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
A 10baseT Ethernet straight through cable is used when the cable is run from a hub to the NL120. A 10baseT Ethernet crossover cable is used if the cable is run directly from the computer to the NL120. For cable length longer than 9 ft, the 10baseT Ethernet cable must be shielded or the 28033 surge suppressor must be used.
The NL120 fastens onto the data logger's 40-pin peripheral port. This means that the NL120 is only compatible with the CR1000 and CR3000. It is NOT compatible with the CR800, CR850, CR200X, CR5000, or CR9000X dataloggers.
Number of FAQs related to NL120: 23
Expand AllCollapse All
The NL120 is a communications peripheral only. It does not provide additional analog terminals for a data logger.
Yes. The NL120 operates as transparent.
Yes, the NL120 can be repaired. Refer to the Repair and Calibration page for the process of returning equipment to Campbell Scientific.
There is no separate firmware for the NL120. It is built into the main data logger’s operating system.
The PakBus protocol allows multiple simultaneous connections. The PakBus address that LoggerNet uses for itself on one of the PCs needs to be changed, though.
No. The NL120 is used for one device only.
Yes. By configuring the NL120 as a CSDC device, both the NL120 and COM320 can share the same CS I/O port. The CR1000 and CR3000 are capable of multi-threading, so the NL120 and COM320 can run in parallel. For more information, refer to the NL120 Ethernet Module Instruction Manual.
The following are some possible causes: