Campbell Scientific’s NL115 enables 10baseT Ethernet communications and stores data on a removable CompactFlash® card. This small, rugged communication device connects to the 40-pin peripheral port on a CR1000 or CR3000 datalogger.Read More
|Power Requirements||12 V supplied through the data logger’s peripheral port|
|Ethernet Speed||10Base-T (10 Mbps), half-duplex|
|Typical Access Speed||200 to 400 kbits s-1|
|Memory Configuration||User selectable; ring (default) or fill-and-stop|
|CF Card Requirements||Industrial-grade|
|Data Logger Operating System (OS)||The CR1000 OS must be OS 9 or later. Both the CR1000 and CR3000 need OS 25 or later to read cards with more than 2 GB of storage.|
|Typical Current Drain||
|Weight||154 g (5.4 oz)|
EMI and ESD Protection
|European Standards||Meets requirements for a class A device under European Standards.|
|Application of Council Directive(s)||89/336/EEC as amended by 89/336/EEC and 93/68/EEC|
|Standards to Which Conformity Is Declared||EN55022-1; 1995 and EN50082-1: 1992|
|LoggerNet||Version 3.2 or later|
|PC400||Version 1.3 or later|
|DevConfig||Version 1.5 or later|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
Campbell Scientific offers and recommends several CF cards (see Ordering tab). These cards have passed Campbell Scientific's ESD testing and operate properly with our data loggers.
Only industrial-grade CF card should be used with our products. Although consumer-grade cards cost less than industrial-grade cards, the consumer-grade cards are more susceptible to failure resulting in both the loss of the card and its stored data. Industrial-grade cards also function over wider temperature ranges and have longer life spans than consumer-grade cards.
The NL115 is set up using the Device Configuration utility (DevConfig). The NL115 requires LoggerNet 3.2 or later, PC400 1.3 or later, and DevConfig 1.5 or later.
The NL115 fastens onto the data logger's 40-pin peripheral port. This means that the NL115 is only compatible with the CR1000 and CR3000. It is NOT compatible with the CR800, CR850, CR200X, CR5000, or CR9000X dataloggers.
To read cards with more than 2 GB of storage, the data logger operating system must be OS 25 or later.
Number of FAQs related to NL115: 31
Expand AllCollapse All
The file on the card is open until the program is stopped or the button on the device is pressed.
No. The CR800 and CR850 do not have the necessary peripheral port to connect to an NL115.
The best practice is to format the CF card, not using “Quick Format” on a PC, before inserting it into a device.
Yes, but the data storage must be managed by the program.
Campbell Scientific always recommends using CF cards carried by Campbell Scientific. Although another CF card might fit and work with a CFM100 or NL115, the CF card might not be rated for the same temperature range as the CFM100 or NL115. In addition, another CF card might not have the same electrostatic discharge (ESD) immunity as the cards tested and carried by Campbell Scientific. For more detail on CF cards, see the “CF Card Information” application note.
For detailed information, review the information in the “CF Card Information” application note.
The CFM100 or NL115 is detected during the data logger boot-up cycle, that is, when it is powered on. Disconnect or connect the CFM100 or NL115 to a data logger only after power to the data logger has been disconnected.
If the CardOut() or TableFile() instructions are configured as fill and stop, the data logger will stop writing data to the CompactFlash cards after they are full.