The AM16/32B multiplexer significantly increases the number of sensors that you can measure with a Campbell Scientific data logger. It interfaces with the data logger and adds terminals so that you can wire additional sensors of almost any type.Read More
Depending on sensor type, the AM16/32B can multiplex 16, 32, or 48 sensors. Up to six AM16/32Bs may be connected to the same data logger, depending on the number of control ports and analog inputs available.
The AM16/32B either multiplexes 16 groups of four lines (a total of 64 lines) through four common (COM) terminals. Alternatively, a manual switch setting allows the AM16/32B to multiplex 32 groups of two lines (also a total of 64 lines) through two COM terminals. A cable connects the common terminals to data logger analog inputs, excitation channels, or ground as required by the sensor. The data logger controls the multiplexer using two control ports, or one control port and one excitation channel.
The equipment is protected from electrical surges using gas tubes on all inputs. A 35 V varistor is connected between the shield ground and ground lug to provide surge protection. The Reset, Clock, and +12 V terminals are protected by 18 V TVS diodes.
|-NOTE-||The protection on the reset, clock, and +12 V inputs are provided by 18 V TVS diodes.|
|Power||Unregulated 9.6 to 16 Vdc|
|Scan Advance||Occurs on the leading edge of the clock pulse transition (from below 1.5 V to above 3.3 V)|
|Minimum Clock Pulse Width||1 ms|
|Maximum Actuation Time for Relay||20 ms|
|Relay Operation||Break before make|
|Initial Relay Resistance, Closed||0.1 ohm|
|Maximum Switching Current||
(Switching currents greater than 30 mA [occasional 50 mA current is acceptable] will degrade the contact surfaces of the mechanical relays and increase their resistance.This will adversely affect the suitability of these relays to multiplex low voltage signals.Although a relay used in this manner no longer qualifies for low voltage measurement, it continues to be useful for switching currents in excess of 30 mA.)
|Maximum Switching Voltage||
A voltage divider such as the VDIV10:1 may be needed between the AM16/32B and the data logger to stay within the input limits of the data logger channel.
|Minimum Contact Life||5 x 107 operations|
|Maximum Contact Voltage Rating||70 V|
|Maximum Voltage||8 Vdc (clock level)|
|Surge||Complies with IEC61000-4-5, test level 3 (±2 kV, 2 ohm coupling impedance)|
|Operating Temperature Range||
|Dimensions||23.9 x 10.2 x 4.6 cm (9.4 x 4.0 x 1.8 in.)|
|Weight||~680 g (~1.5 lb)|
|Air Discharge||Complies with IEC61000-4-2, test level 4 (±15 kV).|
|Contact Discharge||Complies with IEC61000-4-2, test level 4 (±8 kV).|
Typical Current Drain
|Quiescent||< 210 µA|
|Inactive||< 0.9 V|
|Active||3.3 to 8 V|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
|CR300||CRBasic programming experience is recommended, as the AM16/32B with the CR300 series is not currently supported in Short Cut.|
|CR310||CRBasic programming experience is recommended, as the AM16/32B with the CR300 series is not currently supported in Short Cut.|
|CR350||CRBasic programming experience is recommended, as the AM16/32B with the CR350 series is not currently supported in Short Cut.|
|ENC10/12||Mounted sideways only.|
A series of data logger instructions control the AM16/32B operation (measurement instructions that are repetitive are generally contained within a programming loop; in the CR800-series, CR1000, CR3000, and CR5000, this functionality is available in the SubScan instruction).
Compatible sensors include thermistors, potentiometers, load cells, strain gages, vibrating-wire or vibrating-strip transducers, water content reflectometers, and soil moisture blocks. The AM25T is recommended for multiplexing thermocouples.
The maximum number of sensors multiplexed through one AM16/32B depends on the type(s) of sensors measured. For example, assuming identical sensors, the AM16/32B can multiplex up to:
The AM16/32B draws less than 210 microamps quiescent, 6 milliamps active—so power considerations are heavily tied to the percentage of time in quiescent versus active states. Use of a relay address can reduce power consumption by minimizing the time spent in an active state.
In most applications, the data logger’s sealed rechargeable power supply should be more than sufficient; the data logger’s alkaline power supply could be used in applications where the multiplexer is activated infrequently.
The AM16/32B operates in most field conditions but requires a non-condensing environment. A weather-resistant enclosure equipped with desiccant is required for field use. A smaller enclosure is often used if the multiplexer is housed at a distance from the data logger.
If the AM16/32B is to be housed in the data logger’s enclosure, one AM16/32B, a CR800, CR850, CR1000, CR3000, or CR10X datalogger, and a PS150 power supply will fit in an ENC12/14. For convenience in wiring, a larger enclosure (e.g., ENC16/18) is recommended.
LoggerNet, PC400, and ShortCut are compatible with the AM16/32B. Short Cut software (available at no charge from our website) supports simple programming and generates wiring diagrams for AM16/32B applications; early versions of Short Cut did not support multiplexers. Please reference the notes in the compatibility table for data loggers.
Number of FAQs related to AM16/32B: 12
Expand AllCollapse All
Yes. The 3WHB10K is required to measure three-wire temperature probes on the AM16/32B. As both RTDs and thermistors can be measured on the 3WHB10K, refer to the AM16/32B Relay Multiplexer Instruction Manual for the proper programming.
Programming the AM25T multiplexer might be considered more convenient because the newer data loggers have a built-in AM25T CRBasic instruction to handle both AM25T differential thermocouple measurements and differential voltage measurements. There is no CRBasic built-in instruction for the AM16/32B multiplexer. However, the newer versions of the Short Cut program builder software have AM16/32B compatibility for a variety of different sensor types.
The mechanical contact relays of the AM16/32B are subject to corrosion and pitting, which leads to increased relay resistance and greater variations of the resistance with each contact cycle. Switched currents in excess of 30 mA degrade the relay contacts involved, rendering the channel unsuitable for further low-level analog measurements. (If an AM16/32 relay has been used for currents greater than 30 mA, it is not suitable for low-level mV measurements.) The switching of higher level currents causes increased corrosion and pitting of the relays. The solid-state relays on the AM25T multiplexer are not subject to corrosion and pitting; therefore, they are more suitable for low-level mV signals.
The number of switched lines and banks is different between the two multiplexers. The AM16/32B multiplexer has the operating option of running 16 banks of 4 lines each or of running 32 banks of 2 lines each. The AM25T multiplexer has only the one operating mode of 25 banks of 2 lines each. The 4x16 and 2x32 modes of the AM16/32B can offer more channels, convenience, and flexibility in applications than the 2x25-only mode of the AM25T multiplexer can. The 4x16 operation mode of the AM16/32B is a significant advantage for resistive bridge measurements to switch the current to the bridge circuits in addition to lines set up for the bridge output voltage. The 2x32 mode of the AM16/32B and the 2x25 mode of the AM25T multiplexer are commonly used for sensors that require no excitation (that is, voltages and thermocouples). In the 2x32 mode, the AM16/32B multiplexer has seven more channels than the AM25T offers.
The differences between the mechanical relays on the AM16/32B and the solid-state relays on the AM25T are noteworthy with regard to switching time, voltage and current levels, power consumption, and relay activation noise.
Two multiplexers, each with a maximum of 16 or 32 sensors per multiplexer, can be attached. If temperature measurements are excluded from each sensor, 32 sensors are possible.
A practical maximum is to connect one multiplexer per every two control terminals on the data logger. Control terminals can be shared between multiplexers to increase the number of connected multiplexers. Sharing terminals, however, requires more complex wiring and programming. Users who would like to connect more than one multiplexer per every two control terminals are advised to contact a sales or support engineer at Campbell Scientific for assistance.
The mechanical relays on the AM16/32B are subject to wear and thus have a limited lifetime. The minimum life rating of the AM16/32B relays is 107 closures. For a relay cycle interval of 1 second, the minimum lifetime is three years. The solid-state relays on an AM25T don't have this issue.